The PrimeTest 250+ portable appliance tester is battery powered and handheld to allow for fast and efficient testing with clear PASS/FAIL indication and test result values. It has a comprehensive range of mains and battery powered tests, data storage and label printing features.
Newly updated BS EN50699 and 5th Edition compliant products include a test for Class 2 FE equipment, a test for long leads and variable PASS/FAIL limits for both Earth Continuity and Leakage tests.
Enabling simple, professional and fast PAT testing, this versatile tester features all of the required tests of the IET 5th Edition Code of Practice and BS EN50699, along with USB download. It is also fully compatible with the Test n Tag Pro serial printer for quick and easy label printing without any setup required – just plug and print.
Using Seaward’s free datalogger program and pressing a single download key allows test result data to be quickly and easily transferred from the PrimeTest 250+ to PC at the press of a button.
Data is downloaded in a simple CSV spreadsheet format for your PAT records or alternatively it can be imported directly into Seaward’s PATGuard 3 software for more enhanced record keeping, reporting and certification.
Seaward’s unique ‘ZAP’ technology ensures that earth continuity measurements are accurate in situations where other testers struggle to provide reliable measurements, such as measuring through rust, limescale and other surface contamination.
The PrimeTest Series Testers can be used with our free PATMobile App as part of a combined record keeping and reporting solution. The PATMobile App works with the PrimeTest 250+ to enable the input and storage of test records alongside barcode scanning, wireless label printing, photo tagging and mobile reporting.
Built in Britain and used by 1,000’s of organisations every day.
This PAT testing kit includes:
Part number: 403A956
The PrimeTest 250+ Pro Bundle is also available with PATGuard 3 software. Part number: 403A957
The PATMobile accessory bundle has been created to give those using manual PAT Testing equipment the ability to create digital results storage, reporting and labelling solutions when used in conjunction with our PATMobile App
PATMobile is compatible with both IOS and Android
Part number: 403A961
|Output Current||200mA minimum into 2Ω|
|Test Voltage (o/c)||9VDC nominal|
|Test Duration (Class I Test)||2s nominal|
|Display Range||0.01 - 19.99Ω|
|Pass Value (Class I Test)||0.1Ω/0.2Ω/0.3Ω|
|Test Voltage||40V Nominal AC 50Hz Open Circuit|
|Display Range||0.1 - 19.99mA|
|Pass Value||0.5 / 3.5 / 5.0mA|
|Protective Conductor Current|
|Test Voltage||Supply Voltage, max current 13A|
|Test Duration||120s maximum|
|Display Range||0.15 - 9.99mA|
|Pass Value||3.5 / 5.0mA|
|3 phase leakage current (Requires 3 phase adaptor accessory)|
|Test Voltage||3 phase supply voltage|
|Test Duration||Continuous measurement|
|Display Range||0.10 - 9.99mA|
|Power Socket Check|
|Input Voltage Range||225 - 254VAC max current 13A UK|
|Indicates Configuration of Voltage Potential||Line potential phase to earth. Line potential phase to neutral. Line potential neutral to earth|
|Test Voltage||250/500VDC @1mA nomina|
|Test Current||< 2mA (s/c)|
|Test Duration (Class I or II)||2s|
|Display Range||0.01 - 19.99MΩ|
|Pass Value Class I||1.0MΩ|
|Pass Value Class II||2.0MΩ|
|IEC Lead Test|
|Test Voltage||5VDC nominal|
|Test Duration||Time for test (2s max)|
|Test||Live / Neutral Checks for o/c, s/c and crossed|
|Test Voltage||Supply voltage, max current 13A|
|Test Duration||120s maximum|
|Display Range||0.10 - 5mA|
|Pass Value||0.5 / 5.0mA|
|RCD Trip Time|
|RCD Type||Plugtop and fixed installation|
|Test Voltage||Mains supply voltage|
|Test Current||30/150mA rms sinusoidal|
|Test Current Phase||0°/180° auto switched on alternate tests|
|Measuring Range||0 – 500ms|
Register your PrimeTest 250+ today to activate your warranty and to stay up to date with the latest product updates, news and support information.
When carrying out a lead test, the PrimeTest 250+ will test Earth Continuity, Insulation Resistance ...FAQs and Articles
When performing a powered leakage test on the PT250 / PT250+With the power lead connected to the mai...FAQs and Articles
Please see the following instruction:1. Connect your device to the PC using a USB connector.(The ver...FAQs and Articles
To turn on the PT 250+ , press and hold both the class 1 and 2 buttons. ...FAQs and Articles
Sign up to our Newsletter.
Stay up to date with the latest industry and product news, as well as our free educational content such as webinars and our expert guides.