The O/C fault occurs when the polarity test is done. Am I using the correct test for this appliance?
A. The double sockets can be tested using the socket test, and the whole appliance can be tested by means of a Class I test. When performing this test all switches should be turned to their on positions. A continuity test will be done, followed by an insulation resistance test. As the Polarity test would not be included, the O/C fault would not be present.
Sign up to our Newsletter.
Stay up to date with the latest industry and product news, as well as our free educational content such as webinars and our expert guides.