Our offices will be closed on Monday 21st June for a company holiday. We are giving everyone the day off to say thank you for their commitment, dedication and hard work throughout the COVID-19 pandemic.
Our offices will reopen as usual on Tuesday 22nd June.
There is an issue with the csv download of the PT250+, this appears to be with the firmware version 2.70. To permanently fix this it will require a firmware update.
As a temporary measure you can manually correct the download data by:
Deleting “mA” in the “PE Leakage Unit” column
Deleting “Fail” in the “PE Leakage Result” column
The above should be done for both the IEC lead and rcd tests results.