Please click on the link below to find the best way to contact us during the COVID-19 pandemic.Find out more
There is an issue with the csv download of the PT250+, this appears to be with the firmware version 2.70. To permanently fix this it will require a firmware update.
As a temporary measure you can manually correct the download data by:
Deleting “mA” in the “PE Leakage Unit” column
Deleting “Fail” in the “PE Leakage Result” column
The above should be done for both the IEC lead and rcd tests results.