There is an issue with the csv download of the PT250+, this appears to be with the firmware version 2.70. To permanently fix this it will require a firmware update.
As a temporary measure you can manually correct the download data by:
Deleting “mA” in the “PE Leakage Unit” column
Deleting “Fail” in the “PE Leakage Result” column
The above should be done for both the IEC lead and rcd tests results.
Sign up to our Newsletter.
Stay up to date with the latest industry and product news, as well as our free educational content such as webinars and our expert guides.