We have a number of educational webinars for you to view on demandFind out more
Connect the PV200/210 to PC using the USB cable. (This will create a COM port on thePC).
Run the Seaward Solar Datalogger application on the PC. Select the correct COM port.
(Use the Help menu – Trouble shooting guide, to help finding the correct COM port)
Press the Recall key to put the PV200/210 into recall mode, press and hold the Recall key on the PV200/210 while in recall mode. After a few seconds the PV200/210 will transfer all its stored data to the PC.
By default the Datalogger application will save the data in CSV format. This can be opened using Solar Cert or Microsoft Excel. Note that each row of data will have the readings from the Survey 200R where possible.