1. Connect the red and black test leads as shown.
2. Press and hold the Rpe key until the lock icon appears on the display.
3. The resistance between the test probes is displayed.
4. Press the Rpe key to terminate the continuous measurement mode.
Note: Do not connect the PV200/210 to a voltage source while performing the resistance measurement as this will blow the fuse located in the battery compartment.
Note: The continuity measurement is not stored in memory with the automatic test sequence. Continuity measurements are stored as individual records that only record the continuity measurement.