Please click on the link below to find the best way to contact us during the COVID-19 pandemic.Find out more
Tester is detecting more than 1000V through the positive and negative connections, which is more than the tester is rated for. This normally occurs when two strings are connected together, or there is more than one fuse closed in the combiner box. Test button must NOT be pressed. Tester should be disconnected from the system, and a closer inspection of the system carried out visually before re-connecting.